Brand: Everyway Code: acct
US customers click HERE before purchasing, we cannot supply electrical stimulation units or parts to you
We apologise that this product is currently out of stock, COVID has caused some supply issues with stock coming from overseas. We have been informed that new stock has been dispatched and should be here soon. If you would like to be alerted when they arrive please info [at] pelvicfloorexercise.com.au (subject: Please%20inform%20me%20when%20Accurate%20TENS%20is%20back%20in%20stock) (CLICK HERE)
The Accurate or Axion -TENS N103A is a user-friendly portable analogue TENS machine.
TENS can be used for :
- pain relief
- tretament of bladder and bowel urgency and frequency
If you have been prescribed a TENS machine for treating OAB or FI you may find THIS VIDEO helpful.
In the video our Physiotherapist, Fiona, demonstrates how to set up and use your TENS machine for TTNS - Transcutaneous Tibial Nerve Stimulation.
External adhesive electrodes are attached to the inside ankle to gently stimulate the tibial nerve, helping to reduce urinary and faecal urgency.
The Accurate-TENS model N103A allows you to adjust the:
- pulse width (uS)
- frequency (Hz)
- timer ( continuous, 30 min or 60 min)
B : burst : low and high frequency (combination of A-L and Conventional TENS)
C : conventional TENS ( high frequency) with the abiity to control all parameters
M : modulated pulse width
MM : modulated pulse width and frequency
Intensity controls ( on top of machine) and parameter controls (on front of machine) both have clear covers to prevent unexpected changes in any of the settings during use whilst allowing you to see at a glance what settings you are on.
Small, light-weight but durable, the Accurate-TENS is an ideal choice for pain relief and treatment of bladder and bowel urgency.
What you get:
- TENS unit
- 9 volt battery
- 2 x leads
- 1 set of 40 x40mm adhesive electrodes
- instruction manual
- sturdy plastic storage case
This unit will also run on a rechargeable 9V battery, please follow instructions in the manual for use.
CONTRA-INDICATIONS. Do NOT use TENS
- If you have a pacemaker or implanted defibrillator
- If you have an active IPG (implanted pulse generator) - check with your health professional if you can turn off during TENS use
- Over the pregnant uterus
- Over the neck
- If you have epilepsy
- Around actively bleeding tissue
- Around local malignancy
- Near the eyes or testes
- Near an active epiphysis
- In presence of poor circulation or sensation
Dual, isolated between channels
Adjustable, 0-90 mA peak into 500 ohm load each channel
Adjustable, 0-100 Volt (Open Current)
Adjustable, from 2 to 150Hz
Adjustable, from 50 to 300 microseconds
4 Modes, B(Burst), C(Constant), M(Width Modulation), MM(Rate& Width Modulation)
95(L) x 65(W)x23.5(H) mm
Bursts occur twice very second. Pulse width(adjustable), frequency = 100 Hz
The pulse rate and pulse width are adjustable. It generates continuous stimulation based on the setting value.
Modulation Mode(M)Width Modulation
The pulse width is decreased from the preset value by 60% in 3 seconds then back to the original setting. Total cycle time is 6 seconds. Pulse rate(from 2~150Hz) and pulse
width (from 50~300µs) are fully adjustable.
Modulation Mode(MM)Rate & Width Modulation
Modulation mode is a combination of pulse rate and pulse width modulation. The pulse rate and width are automatically varied in a cycle pattern. The pulse width is decreased by 50% from its original setting in 0.5 second, then the pulse rate is decreased by 50% from its original setting in 0.5 second. Total cycle time is 1 second. In this mode, pulse
rate(2-150Hz) and pulse width (50-300µs) are fully adjustable.
Asymmetrical Bi-Phasic Square Pulse
30, 60 minutes or Continue(C)
Max. Charge per pulse
One 9 Volt Battery.(alkaline, or nickel-cadmium rechargeable)
Approximately 50 hours at nominal settings.
Relative Humidity: 30%~75%
Atmosphere Pressure : 700Hpz~1060Hpa
There maybe +/-10% tolerance of all parameters and +/- 20% tolerance of amplitude and voltage